Charles J. Burroughs, Samuel Benz, Todd E. Harvey, H. Sasaki
We have made several improvements to the NIST Josephson FRDC (fast reversed dc) source so that it now operates at 1 Volt. We have made a precision comparison to a conventional semiconductor FRDC source by measuring the thermoelectric transfer difference of a known thermal voltage converter. The agreement between the two sources was measured to be better than 1 part in 10^u7.
Proc., Conference on Precision Electromagnetic Measurements (CPEM)
, Benz, S.
, Harvey, T.
and Sasaki, H.
1 Volt Josephson Fast Reversed DC Source, Proc., Conference on Precision Electromagnetic Measurements (CPEM), Sydney, 1, AS, [online], https://tsapps.nist.gov/publication/get_pdf.cfm?pub_id=10539
(Accessed December 9, 2023)