Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
The International Temperature Scale of 1990 (ITS-90) is defined in terms of "fixed points," i.e. phase transitions of ultra-high-purity materials. We maintain ITS-90 fixed points over the range 83.8058 K to 692.677 K. These comprise triple-point cells of argon (83.8058 K), mercury (234.3156 K), and water (273.16 K); a gallium melting point cell (302.9146 K); and freezing point cells of indium (429.7485 K), tin (505.078 K), and zinc (692.677 K). Each cell is housed in its own thermostat or furnace. In the triple-point cells, liquid, solid, and vapor material are at equilibrium. In the melting and freezing point cells, liquid and solid are in equilibrium with a head space filled with an inert gas (argon) at atmospheric pressure. The uncertainty for these standards ranges from 0.0002 K to 0.002 K. These cells are all portable and, in most cases, are taken to the instrument being calibrated; in this way, the calibration is carried out as a complete system using the same instrumentation, lead wires, etc. as in the actual experiment. In some cases, it is most practical to bring the temperature sensor to the calibration lab.
We also have a standard platinum resistance thermometer (SPRT) that was calibrated by the Temperature Group of NIST-Gaithersburg that is used with a resistance bridge and standard resistor in a "comparison block" calibration furnace for temperatures up to 933 K. The uncertainty for this type of calibration is 0.01 K.
Three piston gage systems with a total of five piston-cylinder sets and 70 kg of standard masses serve as our pressure standards over the range 5 kPa to 70 MPa. A gas-operated system is used for absolute (i.e. relative to vacuum) pressures up to 7 MPa; it has uncertainties of 14 to 20 ppm (parts per million), depending on the piston-cylinder set used. Two hybrid, gas-operated/oil lubricated systems operate at the higher pressures. Our primary high-pressure system measures gage (i.e. relative to atmosphere) pressure and has an uncertainty of 20 ppm. An older system operates up to pressures of 40 MPa with uncertainties of 50 ppm; it is a dual-piston system and has the advantage of also calibrating differential pressure gages.
In experiments requiring the utmost accuracy in pressure we directly connect a piston gage to the fluid property instrument. This approach limits the experiment to inert gases compatible with the piston gage (nitrogen, helium, neon, or argon); it has been used in calibrations with the two-sinker densimeter.
We maintain mass standards ranging from 1 mg to 5 kg. These are standard stainless steel masses of ASTM Class 1 or OIML Class E2. These are regularly sent to the manufacturer's calibration lab for recertification. These are used to calibrate our full range of working laboratory balances, which are used for routine sample preparation, etc. Our most exacting mass determinations (such as recertifying piston gage masses or the sinkers in our magnetic suspension densimeters) are carried out in the calibration lab using one of two "mass comparator" balances. We have a 10 kg comparator with a resolution of 0.1 mg and repeatability of 0.3 mg and a comparator with a capacity of 111 g, resolution of 0.001 mg, and repeatability of 0.003 mg. We also use a 220 g balance with a resolution of 0.01 mg in a "mass comparator mode." The comparators are used together with the standard masses in a double-substitution weighing design [Harris 2003], consisting of four separate weighings: (1) standard masses approximating the mass of the unknown, (2) the unknown object, (3) the unknown object plus a small sensitivity mass, and (4) the standard masses used in weighing 1 plus a sensitivity mass. This design effectively calibrates the balance as an integral part of each weighing. The buoyancy force arising from the density of the ambient air must be taken into account, and this is done by measuring the ambient temperature, pressure, and relative humidity in the balance chamber and applying standard equations for the density of moist air [Davis 1992].
Davis, R.S. (1992). Equation for the determination of the density of moist air (1981/91). Metrologia 29: 67-70.
Harris, G.L. and Torres, J.A. (2003). Selected laboratory and measurement practices and procedures, to support basic mass calibrations. National Institute of Standards and Technology, NISTIR 6969.
Related Programs and Projects:
Intimately related to the issues of calibration is the expression of measurement uncertainty. This is approached in a consistent way at NIST, following Guidelines for Evaluating and Expressing the Uncertainty of NIST Measurement Results:
One of the furnaces that holds the metal melting-point cells (in this case the indium cell at T = 429.748 K; here it is being used to calibrate in-situ the SPRT on the 2-sinker densimeter.
The triple-point of water cell.
Lead Organizational Unit:mml