Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||John R. Stoup; Bryon S. Faust;|
|Title:||Measuring Step Gauges Using the NIST M48 CMM|
|Published:||April 01, 2011|
|Abstract:||The Moore M48 coordinate measuring machine (CMM) has provided NIST with flexible measurement capabilities for many years. The machine‰s state of the art performance was further improved following the move of the machine into the new Advanced Measurement Laboratory (AML) on the Gaithersburg, Maryland site. These improvements are particularly evident in the measurement of 1D step gauges of lengths approaching a meter. This paper will describe the measurement process currently used for these long 1D artifacts. The development of the data collection algorithms, data analysis techniques, and process control methods will be discussed. The impact of thermal issues, elastic deformation resulting from probe contact, gauge fixturing, and coordinate system generation and the impact of these considerations on the measurement uncertainty budget will be presented in detail.|
|Keywords:||CMM, coordinate measuring machine, step gauge, dimensional metrology, uncertainty|
|Research Areas:||Measurements, Measurement Services|
|PDF version:||Click here to retrieve PDF version of paper (464KB)|