Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Paul C. DeRose; Melody V. Smith; Klaus Mielenz; Jeffrey R. Anderson; Gary W. Kramer;|
|Title:||Characterization of Standard Reference Material 2942, Ce-Ion-Doped Glass, Spectral Correction Standard for UV Fluorescence|
|Abstract:||Standard Reference Material (SRM) 2942 is a cuvette-shaped, Ce-ion-doped glass, recommended for use for relative spectral correction of emission and day-to-day performance verification of steady-state fluorescence spectrometers. Properties of this standard that influence its effective use or contribute to the uncertainty in its certified emission spectrum were explored here. These properties include its photostability, absorbance, dissolution rate in water, anisotropy and temperature coefficient of fluorescence intensity. The expanded uncertainties in the certified spectrum are about 9 % around the peak maximum at 330 nm, using an excitation wavelength of 310 nm. The SRM also exhibits a strong resistance to photodegradation, with no measurable decrease in fluorescence intensity even after 25 hours of irradiation with the UV light > 280 nm from a Xe lamp.|
|Citation:||Journal of Luminescence|
|Keywords:||biosystems and health, calibration, fluorescence, instrument qualification, method validation, Ce glass, spectral correction, SRM|
|Research Areas:||Neutron Research, Energy, Manufacturing, Materials Science, Bioscience & Health, Chemistry, Nanotechnology, Physics, Measurements|