Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Jon R. Pratt; David B. Newell; John A. Kramar;|
|Title:||A Flexure Balance With Adjustable Restoring Torque for Nanonewton Force Measurement|
|Published:||January 01, 2002|
|Abstract:||The NIST electrostatic force balance compares mechanical probe forces to an SI realization of force derived from measurements of the capacitance gradient and voltage in an electronic null balance. As we approach the nanonewton regime, the finite stiffness of the guiding flexure in the balance becomes a limiting factor. Here, we consider an equal arm balance mechanism constructed using flexure pivots as an alternative to the compound rectilinear spring of the present design. We review how the sensitivity of such mechanisms can be adjusted either by manipulating the center of gravity, or by applying a negative restoring torque using a spring. We examine tradeoffs associated with tuning the stiffness of these mechanisms with regards to tilt sensitivity and various nonideal flexure behaviors. Results obtained with a prototype mechanism are presented.|
|Conference:||International Measurement Confederation. IMEKO Congress|
|Pages:||pp. 77 - 82|
|Dates:||September 24-26, 2002|
|Research Areas:||Manufacturing, Metrology|