Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Jun-Feng Song; Theodore V. Vorburger;|
|Title:||Standard Deviation or Standard Deviation of the Mean - How to Report Statistical Variation in Surface Calibrations?|
|Published:||October 01, 2004|
|Abstract:||Both the standard deviation and the standard deviation of the mean might be used for reporting the statistical variation in the surface calibration reports. In practice, however, if the calibration laboratories or standard agencies could not provide sufficient information in their calibration reports or standard documents, or the customer might not use this information correctly, the calibration uncertainty might be either over- or under-estimated. Some examples are discussed, suggestions are provided on how to report the statistical variation in standard documents and calibration reports, and how to convert uncertainty values when necessary.|
|Conference:||Proceedings of the American Society for Precision Engineering|
|Proceedings:||Proceedings of American Society for Precision Engineering|
|Dates:||October 26, 2004|
|Keywords:||geometrical non-uniformity,statistical variation,surface calibration,surface roughness,uncertainty|
|Research Areas:||Manufacturing, Metrology|