Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Jodie G. Pope; Christopher J. Crowley;|
|Title:||Removing the Hydrocarbon Liquid from Hydrocarbon Liquid Flow Standards|
|Published:||September 24, 2013|
|Abstract:||NIST and other laboratories calibrate meters that measure the flow of valuable liquids such as petroleum products. Historically, these calibrations have been conducted using Stoddard solvent. Laboratories using Stoddard solvent for liquid-flow calibrations must contend with flammability, inhalation toxicity, and hazardous waste disposal requirements. In response to these health and safety concerns, we modified NISTs piston-prover flow standards so that calibrations are now conducted with mixtures of propylene glycol and water (PG + W). In this paper, we describe the modifications to NIST‰s calibration systems and our experiences in using the modified systems to calibrate turbine, positive displacement, and coriolis meters.|
|Proceedings:||The 16th International Flow Measurement Conference FLOMEKO 2013|
|Dates:||September 24-26, 2013|
|Keywords:||Liquid Flow Calibration, Propylene glycol, Stoddard solvent|
|Research Areas:||Impacts, Measurement Services|
|PDF version:||Click here to retrieve PDF version of paper (727KB)|