Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Stephen Carr; Solomon I. Woods; Timothy M. Jung; Adriaan C. Carter; Raju V. Datla;|
|Title:||PicoWatt Infrared Power Measurement using an Absolute Cryogenic Radiometer|
|Published:||May 07, 2009|
|Abstract:||We report on initial measurements of the low-temperature thermal properties of a device that is similar to the experimental apparatus used for absolute cryogenic radiometry (ACR) within the Low Background Infrared Radiometry (LBIR) facility at NIST. The device consists of a receiver cavity mechanically and thermally connected to a temperature-controlled stage through a thin-walled polyimide tube which serves as a weak thermal link. In order to evaluate the functionality of the device for use in a cryogenic radiometer, we measured the thermal resistance and thermal time constant of the system within the temperature range of 1.8 - 4.4 K. The measured thermal resistance and thermal time constant at 1.883 K were 2400 500 (K/mW) and 24 6 (s). This value for the thermal resistance should result in about an order-of-magnitude increase in radiometer sensitivity compared with the present ACR within LBIR. Although the sensitivity should increase by about an order-of-magnitude, the measured time constant is nearly unchanged with respect to previous ACRs within LBIR, due to the reduced dimensions of the receiver cavity. Finally, the thermal conductivity inferred from the measured thermal resistance and geometrical parameters was computed, with an average value of 0.015 (W/m-K), and compared with other measurements of polyimide from the literature.|
|Proceedings:||SPIE: Defense Security and Sensing 2009|
|Dates:||April 13-17, 2009|
|Keywords:||absolute cryogenic radiometer, electrical substitution radiometer, IR, LBIR, NIST, pW power|
|Research Areas:||Detector Metrology|
|PDF version:||Click here to retrieve PDF version of paper (621KB)|