Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||J O. Cross; W Elam; V G. Harris; J P. Kirkland; Charles E. Bouldin; L B. Sorensen;|
|Title:||Sample-Angle Feedback for Diffraction Anomalous Fine-Structure Spectroscopy|
|Published:||May 01, 1998|
|Abstract:||Diffraction anomalous fine-structure (DAFS) experiments measure Bragg peak intensities as continuous functions of photon energy near a core-level excitation. Measuring the integrated intensity at each energy makes the experiments prohibitively slow, however, in many cases DAFS can be collected quickly by measuring only the peak intensity at the center of the rocking curve. A piezoelectric actuator-driven stage has been designed and tested as part of a sample-angle feedback circuit for locking on to the maximum of the rocking curve while the energy is scanned. Although software peak-tracking requires only a simple calculation of diffractometer angles, it is found that the additional hardware feedback dramatically improves the reproducibility of the data.|
|Citation:||Journal of Synchrotron Radiation|
|Pages:||pp. 911 - 913|
|Research Areas:||Materials Science|