Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Howard C. Reader; Dylan F. Williams; Paul D. Hale; Tracy S. Clement;|
|Title:||Characterization of a 50 GHz Comb Generator|
|Published:||February 01, 2008|
|Abstract:||We characterize a 50 GHz comb generator measured on a sampling oscilloscope. With careful control of the input power, input harmonics and comb generator temperature, we find the output spectrum to be stable to 0.1 dB and 0.5 degrees. We correct results for time-base imperfections before Fourier transforming data into the frequency domain. Thereafter we correct for mismatches, an in-line attenuator, and the complex frequency response of the oscilloscope?s sampler. We discuss the corrected comb generator voltage magnitude and phase delivered to an ideal 50 ? system. We use two sampling heads and report on the comb generator spectrum stability measured over a two week period. We also consider the effect of the recorded time windows, driving power levels, and operating temperatures.|
|Citation:||IEEE Transactions on Microwave Theory and Techniques|
|Pages:||pp. 515 - 521|
|Keywords:||Comb generator stability,mismatch correction,sampling oscilloscope calibration|
|PDF version:||Click here to retrieve PDF version of paper (467KB)|