In June 2007, a spherical integrating source was calibrated in the National Aeronautics and Space Administration(NASA) Goddard Space Flight Centers (GSFC) Calibration Facility as part of the prelaunch characterization program for the NPOESS Preparatory Program (NPP) Ozone Mapping and Profiler Suite (OMPS) instrument. Before shipment to the instrument vendor, the sphere radiance was measured at the Remote Sensing Laboratory at the National Institute of Standards (NIST) and then returned to the NASA Goddard facility for a second calibration. For the NASA GSFC calibration, the reference was a set of quartz halogen lamps procured from NIST. For the measurement in the Remote Sensing Laboratory, the reference was an integrating sphere that was directly calibrated at NISTs Facility for Spectroradiometric Calibrations (FASCAL). For radiances in the visible and near-infrared (400 nm to 1000 nm), the agreement between the NASA GSFC calibration and the validation measurements at the Remote Sensing Laboratory was at the 1 % level. For radiances in the near ultraviolet (250 nm to 400 nm), the agreement was at the 3 % level.
Proceedings Title: Sensor Systems and Next-Generation Satellits XII
Conference Dates: September 16-18, 2008
Conference Location: Cardiff, Wales, UK
Pub Type: Conferences
Calibration, Radiometry, Uncertainty, NPP, OMPS