Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Oliver T. Slattery; Lijun Ma; Paulina S. Kuo; Yong-Su Kim; Xiao Tang;|
|Title:||Tunable up-conversion detector for single photon and bi-photon near-infrared spectroscopic applications|
|Published:||April 29, 2013|
|Abstract:||A tunable waveguide-based frequency up-conversion detector is used for single photon level near infrared (IR) spectroscopic measurements. Applications include direct spectroscopic measurement of week near IR signals and remote bi-photon spectroscopy. We have demonstrated direct spectroscopy of single photon near IR signals from a greatly attenuated laser and a single photon source. We further applied the up-conversion spectrometer for frequency correlated bi-photon spectroscopy using a single photon source of non-degenerate photon pairs at 1310 nm (near IR) and 895 nm. In correlated bi-photon spectroscopy, the spectral function at one wavelength range of a remote object can be reproduced by locally measuring another (near IR) wavelength range using the up-conversion spectrometer and monitoring the coincidence counts. A near IR single photon detection efficiency of 32 % has been achieved with the up-conversion spectrometer. The spectral resolution of the system is approximately 0.2 nm at 1310 nm based on the acceptance width of the up-conversion chip used. In bi-photon spectroscopy, the spectral resolution for the correlated photons at 895 nm is approximately 0.1 nm. The sensitivity achieved using the up-conversion detector is -126 dBm at 1310 nm.|
|Conference:||SPIE Defense, Security + Sensing 2013|
|Proceedings:||Proceedings of SPIE|
|Dates:||April 29-May 3, 2013|
|Keywords:||Correlated bi-photon spectroscopy, up-conversion detector, up-conversion spectrometer|
|Research Areas:||Quantum Information and Measurements|
|PDF version:||Click here to retrieve PDF version of paper (439KB)|