Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Michael P. Unterweger; Ryan P. Fitzgerald;|
|Title:||Technical Note on Unexpected Bias in NIST 4¿¿ Ionization Chamber Measurements|
|Published:||April 04, 2012|
|Abstract:||The NIST 4π pressurized ionization chamber ,AŠ (PIC ,AŠ) has been the mainstay for secondary calibrations of liquid and gaseous gamma-ray emitting sources for the last 40-45 years. It has also been the main instrument used to measure the half-lives of these and similar radionuclide sources. The ionization chamber is the prototype for the TPA Mk.II ionization chamber manufactured by 20th¬ Century Electronics, Ltd. (Centronic), Croydon, England. The chamber and measurement method (Fig. 1) was described in Unterweger et al. (1992) and Calhoun (1987). The basic analysis of the decay data and the chi squared fitting technique are described in Hoppes and Schima (1982). In January of 2010, it was discovered that the position of the source holder (Fig. 2) used for calibrations in the NIST 4π ionization chamber (PIC ,AŠ) has not been constant. The positioning ring that determines the height of the sample in the reentrant tube of the PIC ,AŠ has slowly shifted during 35 years of use. Since the chamber response is height dependent, this|
|Citation:||Applied Radiation and Isotopes|
|Pages:||pp. 1892 - 1893|
|Keywords:||halflife, ionization chamber calibration|
|Research Areas:||Radiation Physics|