Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Gerard N. Stenbakken; D. Liu; J. A. Starzyk; Bryan C. Waltrip;|
|Title:||Nonrandom Quantization Errors in Timebases|
|Published:||May 01, 2000|
|Abstract:||Timebase distortion causes nonlinear distortion of waveforms measured by sampling instruments. When such instruments are used to measure the rms amplitude of the sampled waveforms, such distortions result in errors in the measured rms values. This paper looks at the nature of the errors that result from nonrandom quantization errors in an instrument's timebase circuit. Simulations and measurements on a sampling voltmeter show that the errors in measured rms amplitude have a non-normal probability distribution, such that the probability of large errors is much greater than would be expected from the usual quantization noise model. A novel timebase compensation method is proposed which makes the measured rms errors normally distributed and reduces their standard deviation by a factor of 25. This compensation method was applied to a sampling voltmeter and the improved accuracy was realized.|
|Proceedings:||Proc. IEEE Instrumentation and Technology Conference (IMTC)|
|Pages:||pp. 235 - 240|
|Dates:||May 1-4, 2000|
|Keywords:||correction algorithm,error analysis,quantization,sampling,timebase,voltmeter|
|Research Areas:||Energy Conversion, Storage, and Transport|
|PDF version:||Click here to retrieve PDF version of paper (4MB)|