Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
NIST Authors in Bold
|Author(s):||Tracy S. Clement; Paul D. Hale; Dylan F. Williams; C. M. Wang; Andrew Dienstfrey; Darryl A. Keenan;|
|Title:||Calibration of Sampling Oscilloscopes With High-Speed Photodiodes|
|Published:||August 01, 2006|
|Abstract:||We calibrate the magnitude and phase response of equivalent-time sampling oscilloscopes to 110 GHz. We use a photodiode that has been calibrated with our electrooptic sampling system as a reference input pulse source to the sampling oscilloscope. We account for the impedance of the oscilloscope and the reference photodiode and correct for electrical reflections and distortions due to impedance mismatch. We also correct for time-base imperfections such as drift, time-base distortion, and jitter. We have performed a rigorous uncertainty analysis, which includes a Monte-Carlo simulation of time-domain error sources combined with error sources from the deconvolution of the photodiode pulse, from the mismatch correction, and from the jitter correction.|
|Citation:||IEEE Transactions on Microwave Theory and Techniques|
|Pages:||pp. 3173 - 3181|
|Keywords:||high-speed photodiode,impulse response,mismatch correction,oscilloscope calibration,sampling oscilloscope,uncertainty|
|DOI:||http://dx.doi.org/10.1109/TMTT.2006.879135 (Note: May link to a non-U.S. Government webpage)|