Take a sneak peek at the new NIST.gov and let us know what you think!
(Please note: some content may not be complete on the beta site.).
Please contact the technical staff before shipping instruments or standards to the address listed below.
Fees are subject to change without notice.
The frequency (or, equivalently, the vacuum wavelength) of a red He-Ne laser is determined by comparison to an iodine-stabilized He-Ne laser. The attainable accuracy of this technique is ultimately limited by the uncertainty in the wavelength of the iodine stabilized laser. The relative standard uncertainty of the laser vacuum wavelength at NIST is 5 x 10-11 . The relative expanded uncertainty is 1 x 10-10 . The relative expanded uncertainty can be reduced to 5 x 10-11 on special requests. As a practical matter, however, the uncertainty of a comparison is usually determined by the limited reproducibility of the laser under test.